[vc_row css=".vc_custom_1477037811368{margin-bottom: 25px !important;}"][vc_column width="1/4" css=".vc_custom_1477040108336{padding-top: 12px !important;}"][vc_column_text css=".vc_custom_1477667674582{margin-bottom: 15px !important;}"]
<h4 class="title">Noise & Vibration Products:</h4>
[/vc_column_text][vc_wp_custommenu nav_menu="97" el_class="product-side-nav"][/vc_column][vc_column width="3/4"][vc_row_inner][vc_column_inner width="7/12"][vc_column_text]
<h5 class="title">i-dB Noise Monitor</h5>
<p class="subtitle">i-dB is Turnkey’s specially-developed noise monitor. It offers users a simple and accurate approach to monitoring noise in a range of environments.</p>
IEC 61672-1 Class 2 comes as standard, and IEC 61672-1 Class 1 is available for those requiring a higher degree of accuracy.[/vc_column_text][/vc_column_inner][vc_column_inner width="5/12"][vc_single_image image="13543" img_size="full" css_animation="right-to-left"][/vc_column_inner][/vc_row_inner][vc_column_text]The i-dB has a range of uses, and its versatility makes it a suitable choice for monitoring noise in industrial settings, on construction sites and at sporting events and music venues, as well as measuring traffic noise.

Compact and housed in an IP65 enclosure, the i-dB is protected from dust and water, making it sturdy and reliable. It performs in all weathers and conditions, delivering accurate noise readings in temperatures from -10 °C to +50 °C, and in humidity up to 95%.

With a measurement range of 30 dB to 120 dB RMS the i-dB will record noise in commonly used parameter such as LAeq, LAmax, LA10, LA90 (as well as other options).

Designed and developed in-house, the i-dB is compatible with our Osiris or Topas dust and air pollution monitoring instruments. Used together, they provide a complete solution for the constant monitoring of noise levels and air quality.

With built-in internet connectivity, the i-dB works with our AirQWeb online interface. This data is transferred and AirQWeb provides a free, user-friendly platform for accessing, reviewing, and storing your data. It enables users to generate graphs and reports, which can be converted to Excel or PDF documents for further reporting and analysis.

All results can be recorded online using our free online application, AirQWeb. We have also developed the AirQApp for Android and Apple mobile devices, enabling users to access data on the go. With the app, users can generate 3D graphs, start and stop sampling, change the configuration of the sensors, and generate reports.
[/vc_column_text][/vc_column][/vc_row]